12/14/23 12:41:23.323	Information	Image Studio Version 5.2.5	GMT -5:00
12/14/23 12:41:23.323	Information	OS: Windows 8 6.2	GMT -5:00
12/14/23 12:41:23.339	Information	Java Home: c:\program files\licor\image studio\jre	GMT -5:00
12/14/23 12:41:23.339	Information	JRE Version: 1.7.0_72 32-bit	GMT -5:00
12/14/23 12:41:23.339	Information	Java library: C:\Program Files\Licor\Image Studio;C:\WINDOWS\Sun\Java\bin;C:\WINDOWS\system32;C:\WINDOWS;C:\Program Files (x86)\Common Files\Oracle\Java\javapath;C:\WINDOWS\system32;C:\WINDOWS;C:\WINDOWS\System32\Wbem;C:\WINDOWS\System32\WindowsPowerShell\v1.0\;C:\WINDOWS\System32\OpenSSH\;C:\Program Files\PolicyPak\;C:\Users\wmao2\AppData\Local\Microsoft\WindowsApps;;c:\program files\licor\image studio\jre\bin;. 32-bit	GMT -5:00
12/14/23 12:41:23.448	Information	Database migrations: 5.0_Initial, 5.0.66_New log entries fks, 5.0.67_Change lane wrappers, 5.2.68_Update database version, 5.2.69_Add Image Id Constraint	GMT -5:00
12/14/23 12:41:23.448	Information	Installed Analysis Key: Grid_Analysis	GMT -5:00
12/14/23 12:41:23.448	Information	Installed Analysis Key: Image_Studio_Base	GMT -5:00
12/14/23 12:41:23.464	Information	Installed Analysis Key: Shapes	GMT -5:00
12/14/23 12:41:23.464	Information	Installed Analysis Key: Western_Analysis	GMT -5:00
12/14/23 12:41:23.464	Information	Installed Instrument Key: CDG_Acquire	GMT -5:00
12/14/23 12:41:23.464	Information	Installed Instrument Key: Odyssey_Acquire	GMT -5:00
12/14/23 12:41:23.464	Information	Installed Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/14/23 12:41:23.479	Information	Installed Instrument Key: OdysseyFc_Acquire	GMT -5:00
12/14/23 12:41:23.479	Information	Installed Instrument Key: OdysseySa_Acquire	GMT -5:00
12/14/23 12:41:23.479	Information	Installed Instrument Key: Pearl_Acquire	GMT -5:00
12/14/23 12:41:23.479	Information	Selected Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/14/23 12:41:23.479	Information	Loading project C:\Users\wmao2\2023-12-14\Images\...	GMT -5:00
12/14/23 12:41:23.589	Information	Loading more acquisitions...	GMT -5:00
12/14/23 12:41:24.042	Information	Finished adding Acquisitions to Work Area.	GMT -5:00
12/14/23 12:41:24.057	Information	Image Studio Version 5.2.5	GMT -5:00
12/14/23 12:41:24.448	Information	USBCat Client Version 1.0.2	GMT -5:00
12/14/23 12:41:30.665	Information	Have discovered 1 instruments.	GMT -5:00
12/14/23 12:41:30.665	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/14/23 12:41:30.665	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/14/23 12:41:30.681	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/14/23 12:41:30.728	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/14/23 12:41:30.775	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/14/23 12:41:31.930	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    653 mV 	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/14/23 12:41:33.758	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                31 C  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/14/23 12:41:33.774	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/14/23 12:41:33.805	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/14/23 12:41:45.412	Information	[Instrument Request]  1 ScanAutoEstimate 169um lowest 0.0 0.1 5.3 0.0 4.9 700 800 	GMT -5:00
12/14/23 12:41:45.427	Information	[Instrument Response] 1 Status 0 Executing command: ScanAutoEstimate 169um lowest 0 0.1 5.3 0 4.9 700 800	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 X mapping: [x0=0.1,x1=5.3] => [left=-13,right=2459] w=308	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Seconds/line=0.236549	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 linesPerBlock=42 flushLinesPerBlock=21	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Blocks=15	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Status 0 Estimates (sec): setup=2.02 line=0.250742 => total=148	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Progress 0% 3 minutes	GMT -5:00
12/14/23 12:41:45.474	Information	[Instrument Response] 1 Result 0 OK	GMT -5:00
12/14/23 12:41:51.239	Information	[Instrument Request]  2 ScanAutoEstimate 169um medium 0.0 0.1 5.3 0.0 4.9 700 800 	GMT -5:00
12/14/23 12:41:51.270	Information	[Instrument Response] 2 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5.3 0 4.9 700 800	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 X mapping: [x0=0.1,x1=5.3] => [left=-13,right=2459] w=308	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Seconds/line=0.350775	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 linesPerBlock=29 flushLinesPerBlock=14	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Blocks=21	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Status 0 Estimates (sec): setup=2.02 line=0.371821 => total=218	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:41:51.301	Information	[Instrument Response] 2 Result 0 OK	GMT -5:00
12/14/23 12:41:57.659	Information	Starting acquisition...	GMT -5:00
12/14/23 12:41:59.096	Information	[Instrument Request]  3 ScanAuto 169um medium 0.0 0.1 5.3 0.0 4.9 700 800 	GMT -5:00
12/14/23 12:41:59.221	Information	[Instrument Response] 3 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5.3 0 4.9 700 800	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 X mapping: [x0=0.1,x1=5.3] => [left=-13,right=2459] w=308	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Seconds/line=0.350775	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 linesPerBlock=29 flushLinesPerBlock=14	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Blocks=21	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Status 0 Estimates (sec): setup=2.02 line=0.371821 => total=218	GMT -5:00
12/14/23 12:41:59.237	Information	[Instrument Response] 3 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:41:59.253	Information	[Instrument Response] 3 Progress 0% Initializing	GMT -5:00
12/14/23 12:41:59.268	Information	[Instrument Response] 3 Status 0 Preparation complete	GMT -5:00
12/14/23 12:42:02.111	Information	[Instrument Response] 3 Status 0 Motors initialized	GMT -5:00
12/14/23 12:42:02.158	Information	[Instrument Response] 3 Status 0 X-Motor configured	GMT -5:00
12/14/23 12:42:02.158	Information	[Instrument Response] 3 Status 0 DSP configured	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Digital / DSP configured	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Cooling	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 700 20 0 0 308 14 308 290 xofs=1 yofs=0	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 800 20 0 0 308 14 308 290 xofs=0 yofs=0	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=14 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=29 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/14/23 12:42:02.205	Information	[Instrument Response] 3 Status 0 Move motors to initial positions	GMT -5:00
12/14/23 12:42:02.861	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/14/23 12:42:02.924	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:42:04.876	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:42:06.470	Information	[Instrument Response] 3 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:42:06.876	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:42:06.938	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:42:06.938	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 12:42:06.938	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:42:06.938	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:42:06.938	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:42:08.875	Information	[Instrument Response] 3 Progress 1% 5 minutes	GMT -5:00
12/14/23 12:42:09.250	Information	[Instrument Response] 3 Progress 1% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:42:11.906	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/14/23 12:42:14.905	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:42:14.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/14/23 12:42:14.968	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:42:14.968	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:42:15.921	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:42:17.920	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:42:18.920	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:42:21.419	Information	[Instrument Response] 3 ImageBlock 700 0 0 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:21.419	Information	[Instrument Response] 3 ImageBlock 800 0 0 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:21.435	Information	Transferring image block at (0,0) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:42:21.482	Information	Transferring image block at (0,0) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:42:22.950	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/14/23 12:42:27.981	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         106 V  	GMT -5:00
12/14/23 12:42:28.027	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:42:28.027	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:42:28.027	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:42:28.980	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:42:34.807	Information	[Instrument Response] 3 ImageBlock 700 0 14 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:34.807	Information	[Instrument Response] 3 ImageBlock 800 0 14 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:34.823	Information	Transferring image block at (0,14) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:42:34.854	Information	Transferring image block at (0,14) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:42:38.384	Information	[Instrument Response] 3 Progress 11% 4 minutes	GMT -5:00
12/14/23 12:42:41.024	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:42:41.087	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/14/23 12:42:41.087	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/14/23 12:42:41.087	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:42:41.087	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:42:41.087	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:42:42.040	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:42:42.087	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 12:42:42.087	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:42:43.040	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:42:47.055	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:42:47.805	Information	[Instrument Response] 3 ImageBlock 700 0 28 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:47.805	Information	[Instrument Response] 3 ImageBlock 800 0 28 308 14 308 290 8624	GMT -5:00
12/14/23 12:42:47.805	Information	Transferring image block at (0,28) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:42:47.852	Information	Transferring image block at (0,28) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:42:48.055	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 12:42:55.100	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/14/23 12:42:55.147	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:42:55.147	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 12:42:55.147	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:42:56.100	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:43:01.193	Information	[Instrument Response] 3 ImageBlock 700 0 42 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:01.193	Information	[Instrument Response] 3 ImageBlock 800 0 42 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:01.193	Information	Transferring image block at (0,42) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:43:01.224	Information	Transferring image block at (0,42) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:43:07.519	Information	[Instrument Response] 3 Progress 22% 3 minutes, 30 seconds	GMT -5:00
12/14/23 12:43:08.144	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:43:08.207	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/14/23 12:43:08.207	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:43:08.207	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:43:09.144	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:43:14.190	Information	[Instrument Response] 3 ImageBlock 700 0 56 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:14.190	Information	[Instrument Response] 3 ImageBlock 800 0 56 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:14.190	Information	Transferring image block at (0,56) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:43:14.221	Information	Transferring image block at (0,56) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:43:16.174	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:43:17.189	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:43:21.204	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/14/23 12:43:21.251	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:43:21.251	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:43:21.251	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:43:21.251	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:43:21.251	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:43:22.204	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 12:43:22.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:43:22.250	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 12:43:27.577	Information	[Instrument Response] 3 ImageBlock 700 0 70 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:27.577	Information	[Instrument Response] 3 ImageBlock 800 0 70 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:27.593	Information	Transferring image block at (0,70) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:43:27.609	Information	Transferring image block at (0,70) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:43:35.263	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:43:35.341	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/14/23 12:43:35.341	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:43:35.341	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:43:36.263	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:43:38.262	Information	[Instrument Response] 3 Progress 33% 3 minutes	GMT -5:00
12/14/23 12:43:40.574	Information	[Instrument Response] 3 ImageBlock 700 0 84 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:40.574	Information	[Instrument Response] 3 ImageBlock 800 0 84 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:40.590	Information	Transferring image block at (0,84) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:43:40.606	Information	Transferring image block at (0,84) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:43:45.308	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:43:46.308	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:43:48.307	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         106 V  	GMT -5:00
12/14/23 12:43:48.370	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:43:48.370	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:43:48.370	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:43:49.323	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:43:50.322	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/14/23 12:43:53.962	Information	[Instrument Response] 3 ImageBlock 700 0 98 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:53.962	Information	[Instrument Response] 3 ImageBlock 800 0 98 308 14 308 290 8624	GMT -5:00
12/14/23 12:43:53.978	Information	Transferring image block at (0,98) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:43:53.993	Information	Transferring image block at (0,98) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:43:56.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:43:57.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:44:00.367	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          55 mA 	GMT -5:00
12/14/23 12:44:01.367	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:44:01.429	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/14/23 12:44:01.429	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:44:01.429	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:44:01.429	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:44:02.367	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 12:44:02.429	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:44:03.366	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:44:06.975	Information	[Instrument Response] 3 ImageBlock 700 0 112 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:06.975	Information	[Instrument Response] 3 ImageBlock 800 0 112 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:06.990	Information	Transferring image block at (0,112) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:44:07.006	Information	Transferring image block at (0,112) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:44:07.397	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:44:08.396	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:44:08.584	Information	[Instrument Response] 3 Progress 45% 2 minutes, 30 seconds	GMT -5:00
12/14/23 12:44:15.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/14/23 12:44:15.473	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:44:15.473	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:44:15.473	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:44:16.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:44:18.441	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:44:19.441	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:44:20.378	Information	[Instrument Response] 3 ImageBlock 700 0 126 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:20.378	Information	[Instrument Response] 3 ImageBlock 800 0 126 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:20.394	Information	Transferring image block at (0,126) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:44:20.425	Information	Transferring image block at (0,126) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:44:27.471	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:44:28.471	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:44:28.534	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/14/23 12:44:28.534	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:44:28.534	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:44:28.534	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:44:29.471	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:44:33.398	Information	[Instrument Response] 3 ImageBlock 700 0 140 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:33.398	Information	[Instrument Response] 3 ImageBlock 800 0 140 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:33.398	Information	Transferring image block at (0,140) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:44:33.429	Information	Transferring image block at (0,140) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:44:38.552	Information	[Instrument Response] 3 Progress 56% 2 minutes	GMT -5:00
12/14/23 12:44:40.520	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:44:41.535	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         131 V  	GMT -5:00
12/14/23 12:44:41.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:44:41.582	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:44:41.582	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:44:42.535	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/14/23 12:44:42.582	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:44:43.535	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:44:46.816	Information	[Instrument Response] 3 ImageBlock 700 0 154 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:46.816	Information	[Instrument Response] 3 ImageBlock 800 0 154 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:46.816	Information	Transferring image block at (0,154) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:44:46.847	Information	Transferring image block at (0,154) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:44:48.425	Information	[Instrument Response] 3 Progress 60% 1 minute, 50 seconds	GMT -5:00
12/14/23 12:44:51.566	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:44:52.566	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:44:55.581	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:44:55.643	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:44:55.643	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:44:55.643	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:44:58.707	Information	[Instrument Response] 3 Progress 63% 1 minute, 40 seconds	GMT -5:00
12/14/23 12:44:59.847	Information	[Instrument Response] 3 ImageBlock 700 0 168 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:59.847	Information	[Instrument Response] 3 ImageBlock 800 0 168 308 14 308 290 8624	GMT -5:00
12/14/23 12:44:59.847	Information	Transferring image block at (0,168) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:44:59.879	Information	Transferring image block at (0,168) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:45:08.168	Information	[Instrument Response] 3 Progress 67% 1 minute, 30 seconds	GMT -5:00
12/14/23 12:45:08.637	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         115 V  	GMT -5:00
12/14/23 12:45:08.684	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:45:08.684	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:45:08.684	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:45:09.637	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:45:13.280	Information	[Instrument Response] 3 ImageBlock 700 0 182 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:13.280	Information	[Instrument Response] 3 ImageBlock 800 0 182 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:13.280	Information	Transferring image block at (0,182) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:45:13.312	Information	Transferring image block at (0,182) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:45:15.655	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:45:15.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:45:16.671	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:45:16.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:45:18.842	Information	[Instrument Response] 3 Progress 71% 1 minute, 20 seconds	GMT -5:00
12/14/23 12:45:21.681	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/14/23 12:45:21.744	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         145 V  	GMT -5:00
12/14/23 12:45:21.744	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:45:21.744	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:45:21.744	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:45:22.696	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          85 V  	GMT -5:00
12/14/23 12:45:22.743	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:45:23.696	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:45:26.321	Information	[Instrument Response] 3 ImageBlock 700 0 196 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:26.321	Information	[Instrument Response] 3 ImageBlock 800 0 196 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:26.337	Information	Transferring image block at (0,196) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:45:26.384	Information	Transferring image block at (0,196) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:45:28.712	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:45:28.775	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:45:28.775	Information	[Instrument Response] 3 Progress 75% 1 minute, 10 seconds	GMT -5:00
12/14/23 12:45:29.712	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:45:29.775	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:45:35.747	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/14/23 12:45:35.794	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:45:35.794	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:45:35.794	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:45:36.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:45:38.625	Information	[Instrument Response] 3 Progress 78% 1 minute	GMT -5:00
12/14/23 12:45:39.765	Information	[Instrument Response] 3 ImageBlock 700 0 210 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:39.765	Information	[Instrument Response] 3 ImageBlock 800 0 210 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:39.765	Information	Transferring image block at (0,210) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:45:39.797	Information	Transferring image block at (0,210) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:45:41.781	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:45:42.780	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:45:48.484	Information	[Instrument Response] 3 Progress 82% 50 seconds	GMT -5:00
12/14/23 12:45:48.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:45:48.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    704 mV 	GMT -5:00
12/14/23 12:45:48.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         125 V  	GMT -5:00
12/14/23 12:45:48.859	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:45:48.859	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:45:49.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:45:49.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 12:45:50.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:45:52.816	Information	[Instrument Response] 3 ImageBlock 700 0 224 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:52.816	Information	[Instrument Response] 3 ImageBlock 800 0 224 308 14 308 290 8624	GMT -5:00
12/14/23 12:45:52.816	Information	Transferring image block at (0,224) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:45:52.847	Information	Transferring image block at (0,224) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:45:54.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:45:55.832	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:45:58.769	Information	[Instrument Response] 3 Progress 86% 40 seconds	GMT -5:00
12/14/23 12:46:02.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/14/23 12:46:02.906	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:46:02.906	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:46:02.906	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:46:03.863	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:46:05.878	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:46:06.253	Information	[Instrument Response] 3 ImageBlock 700 0 238 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:06.253	Information	[Instrument Response] 3 ImageBlock 800 0 238 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:06.253	Information	Transferring image block at (0,238) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:46:06.284	Information	Transferring image block at (0,238) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:46:06.877	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:46:08.662	Information	[Instrument Response] 3 Progress 89% 30 seconds	GMT -5:00
12/14/23 12:46:15.910	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:46:15.958	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/14/23 12:46:15.958	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:46:15.958	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:46:16.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:46:18.923	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:46:18.970	Information	[Instrument Response] 3 Progress 93% 20 seconds	GMT -5:00
12/14/23 12:46:19.298	Information	[Instrument Response] 3 ImageBlock 700 0 252 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:19.298	Information	[Instrument Response] 3 ImageBlock 800 0 252 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:19.313	Information	Transferring image block at (0,252) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:46:19.344	Information	Transferring image block at (0,252) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:46:19.923	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:46:28.812	Information	[Instrument Response] 3 Progress 97% 10 seconds	GMT -5:00
12/14/23 12:46:28.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         137 V  	GMT -5:00
12/14/23 12:46:29.015	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:46:29.015	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:46:29.015	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:46:29.015	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:46:29.968	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          88 V  	GMT -5:00
12/14/23 12:46:30.030	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:46:30.967	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:46:31.983	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:46:32.733	Information	[Instrument Response] 3 ImageBlock 700 0 266 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:32.733	Information	[Instrument Response] 3 ImageBlock 800 0 266 308 14 308 290 8624	GMT -5:00
12/14/23 12:46:32.748	Information	Transferring image block at (0,266) with size 308x14 to the 700 channel.	GMT -5:00
12/14/23 12:46:32.764	Information	Transferring image block at (0,266) with size 308x14 to the 800 channel.	GMT -5:00
12/14/23 12:46:32.983	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:46:36.732	Information	[Instrument Response] 3 Progress 100% 0 seconds	GMT -5:00
12/14/23 12:46:36.997	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:46:37.060	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/14/23 12:46:37.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:46:37.060	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:46:37.997	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 12:46:38.731	Information	[Instrument Response] 3 ImageBlock 700 0 280 308 10 308 290 6160	GMT -5:00
12/14/23 12:46:38.731	Information	[Instrument Response] 3 ImageBlock 800 0 280 308 10 308 290 6160	GMT -5:00
12/14/23 12:46:38.731	Information	Transferring image block at (0,280) with size 308x10 to the 700 channel.	GMT -5:00
12/14/23 12:46:38.747	Information	[Instrument Response] 3 Status 0 COMPLETE SCAN 272.246 seconds	GMT -5:00
12/14/23 12:46:38.747	Information	Transferring image block at (0,280) with size 308x10 to the 800 channel.	GMT -5:00
12/14/23 12:46:38.809	Information	[Instrument Response] 3 Status 0 Averages (sec): setup=2.00005 line=0.393526 => total=231	GMT -5:00
12/14/23 12:46:38.809	Information	[Instrument Response] 3 Status 0 Comparisons: setup=1.00997 line=0.944845	GMT -5:00
12/14/23 12:46:38.809	Information	[Instrument Response] 3 Status 0 Finalize motors	GMT -5:00
12/14/23 12:46:38.809	Information	[Instrument Response] 3 Progress 100% Scan complete	GMT -5:00
12/14/23 12:46:38.841	Information	[Instrument Response] 3 Result 0 OK	GMT -5:00
12/14/23 12:46:38.841	Information	Acquisition 0000001_01 finished with channels: 700 800.	GMT -5:00
12/14/23 12:46:39.747	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/14/23 12:46:39.762	Information	Scan completed successfully	GMT -5:00
12/14/23 12:46:39.872	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         131 V  	GMT -5:00
12/14/23 12:46:39.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          83 V  	GMT -5:00
12/14/23 12:46:39.887	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 12:46:39.903	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 12:46:39.903	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 12:46:40.012	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          78 V  	GMT -5:00
12/14/23 12:46:40.059	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          49 V  	GMT -5:00
12/14/23 12:46:41.012	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          47 V  	GMT -5:00
12/14/23 12:46:41.075	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          30 V  	GMT -5:00
12/14/23 12:46:42.019	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          31 V  	GMT -5:00
12/14/23 12:46:42.079	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          19 V  	GMT -5:00
12/14/23 12:46:43.029	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          21 V  	GMT -5:00
12/14/23 12:46:43.074	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/14/23 12:46:44.029	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/14/23 12:46:44.079	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/14/23 12:46:45.035	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/14/23 12:46:45.089	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/14/23 12:46:46.038	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/14/23 12:46:47.057	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/14/23 12:46:47.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/14/23 12:46:49.047	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/14/23 12:46:50.049	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      234 mA 	GMT -5:00
12/14/23 12:46:51.056	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:46:53.061	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:48:34.875	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/14/23 12:49:25.310	Information	Starting preview acquisition...	GMT -5:00
12/14/23 12:49:26.201	Information	[Instrument Request]  4 ScanAuto preview lowest 0.0 0.1 5.3 0.0 4.9 700 800 	GMT -5:00
12/14/23 12:49:40.619	Information	Canceling acquisition...	GMT -5:00
12/14/23 12:49:40.619	Information	[Instrument Request]  5 Stop 	GMT -5:00
12/14/23 12:51:40.624	Error      	The 'Stop' command timed out. The elapsed time exceeded 120 seconds. Check the instrument connection.	GMT -5:00
12/14/23 12:53:26.218	Error      	The 'ScanAuto preview lowest 0.0 0.1 5.3 0.0 4.9 700 800' command timed out. The elapsed time exceeded 240 seconds. Check the instrument connection.	GMT -5:00
12/14/23 12:53:45.811	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/14/23 12:54:00.166	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/14/23 12:54:54.054	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/14/23 12:54:59.041	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/14/23 12:54:59.088	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/14/23 12:54:59.135	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/14/23 12:54:59.151	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/14/23 12:54:59.151	Information	[Instrument Request]  6 ScanAutoEstimate 169um medium 0.0 0.1 5.3 0.0 4.9 700 800 	GMT -5:00
12/14/23 12:55:02.107	Information	[Instrument Response] 6 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5.3 0 4.9 700 800	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 X mapping: [x0=0.1,x1=5.3] => [left=-13,right=2459] w=308	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Seconds/line=0.350775	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 linesPerBlock=29 flushLinesPerBlock=14	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Blocks=21	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Status 0 Estimates (sec): setup=2.02 line=0.371821 => total=218	GMT -5:00
12/14/23 12:55:02.122	Information	[Instrument Response] 6 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 6 Result 0 OK	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 4.98 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/14/23 12:55:02.138	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/14/23 12:55:02.169	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/14/23 12:55:17.347	Information	[Instrument Request]  7 ScanAutoEstimate 169um medium 0.0 0.0 5.7 0.0 5.1 700 800 	GMT -5:00
12/14/23 12:55:17.378	Information	[Instrument Response] 7 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 5.7 0 5.1 700 800	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 X mapping: [x0=0,x1=5.7] => [left=-60,right=2644] w=337	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Seconds/line=0.375775	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 linesPerBlock=27 flushLinesPerBlock=13	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Blocks=23	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Status 0 Estimates (sec): setup=2.02 line=0.398321 => total=243	GMT -5:00
12/14/23 12:55:17.409	Information	[Instrument Response] 7 Progress 0% 5 minutes	GMT -5:00
12/14/23 12:55:17.425	Information	[Instrument Response] 7 Result 0 OK	GMT -5:00
12/14/23 12:55:19.948	Information	Starting acquisition...	GMT -5:00
12/14/23 12:55:20.916	Information	[Instrument Request]  8 ScanAuto 169um medium 0.0 0.0 5.7 0.0 5.1 700 800 	GMT -5:00
12/14/23 12:55:20.979	Information	[Instrument Response] 8 Status 1 Executing command: ScanAuto 169um medium 0 0 5.7 0 5.1 700 800	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 X mapping: [x0=0,x1=5.7] => [left=-60,right=2644] w=337	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Computed scan parameters	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Seconds/line=0.375775	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 linesPerBlock=27 flushLinesPerBlock=13	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Blocks=23	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Status 0 Estimates (sec): setup=2.02 line=0.398321 => total=243	GMT -5:00
12/14/23 12:55:20.994	Information	[Instrument Response] 8 Progress 0% 5 minutes	GMT -5:00
12/14/23 12:55:21.010	Information	[Instrument Response] 8 Progress 0% Initializing	GMT -5:00
12/14/23 12:55:21.026	Information	[Instrument Response] 8 Status 0 Preparation complete	GMT -5:00
12/14/23 12:55:23.929	Information	[Instrument Response] 8 Status 0 Motors initialized	GMT -5:00
12/14/23 12:55:23.992	Information	[Instrument Response] 8 Status 0 X-Motor configured	GMT -5:00
12/14/23 12:55:23.992	Information	[Instrument Response] 8 Status 0 DSP configured	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Digital / DSP configured	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Cooling	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Create MergeImageChannel: 700 20 0 0 337 13 337 302 xofs=1 yofs=0	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Create MergeImageChannel: 800 20 0 0 337 13 337 302 xofs=0 yofs=0	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 start=34 scan_lines=13 line=0 end=302 700=Low,L1 800=High,4	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 start=34 scan_lines=27 line=0 end=302 700=High,4 800=Low,L1	GMT -5:00
12/14/23 12:55:24.039	Information	[Instrument Response] 8 Status 0 Move motors to initial positions	GMT -5:00
12/14/23 12:55:24.195	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/14/23 12:55:24.257	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:55:25.657	Information	[Instrument Response] 8 Progress 0% 5 minutes	GMT -5:00
12/14/23 12:55:26.204	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:55:26.266	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:55:26.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:55:26.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:55:26.266	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:55:32.235	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/14/23 12:55:32.281	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:55:32.281	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:55:33.234	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/14/23 12:55:33.297	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         139 V  	GMT -5:00
12/14/23 12:55:33.297	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:55:33.297	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:55:34.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          87 V  	GMT -5:00
12/14/23 12:55:34.291	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:55:34.291	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:55:35.245	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:55:40.478	Information	[Instrument Response] 8 ImageBlock 700 0 0 337 13 337 302 8762	GMT -5:00
12/14/23 12:55:40.478	Information	[Instrument Response] 8 ImageBlock 800 0 0 337 13 337 302 8762	GMT -5:00
12/14/23 12:55:40.493	Information	Transferring image block at (0,0) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:55:40.509	Information	Transferring image block at (0,0) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:55:44.275	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:55:45.290	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:55:47.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/14/23 12:55:47.351	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:55:47.351	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:55:47.351	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:55:48.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:55:51.307	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 12:55:51.370	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:55:52.308	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:55:52.370	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:55:53.776	Information	[Instrument Response] 8 ImageBlock 700 0 13 337 13 337 302 8762	GMT -5:00
12/14/23 12:55:53.776	Information	[Instrument Response] 8 ImageBlock 800 0 13 337 13 337 302 8762	GMT -5:00
12/14/23 12:55:53.791	Information	Transferring image block at (0,13) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:55:53.838	Information	Transferring image block at (0,13) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:55:56.332	Information	[Instrument Response] 8 Progress 10% 4 minutes, 30 seconds	GMT -5:00
12/14/23 12:55:57.332	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      305 mA 	GMT -5:00
12/14/23 12:56:00.340	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:56:00.403	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/14/23 12:56:00.403	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:56:00.403	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:56:01.340	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:56:06.658	Information	[Instrument Response] 8 ImageBlock 700 0 26 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:06.658	Information	[Instrument Response] 8 ImageBlock 800 0 26 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:06.658	Information	Transferring image block at (0,26) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:56:06.689	Information	Transferring image block at (0,26) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:56:13.390	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         116 V  	GMT -5:00
12/14/23 12:56:13.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:56:13.453	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:56:13.453	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:56:14.406	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:56:19.936	Information	[Instrument Response] 8 ImageBlock 700 0 39 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:19.936	Information	[Instrument Response] 8 ImageBlock 800 0 39 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:19.952	Information	Transferring image block at (0,39) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:56:19.967	Information	Transferring image block at (0,39) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:56:26.257	Information	[Instrument Response] 8 Progress 20% 4 minutes	GMT -5:00
12/14/23 12:56:26.445	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/14/23 12:56:26.507	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         135 V  	GMT -5:00
12/14/23 12:56:26.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:56:26.507	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:56:26.507	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:56:27.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/14/23 12:56:27.507	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:56:28.460	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:56:32.819	Information	[Instrument Response] 8 ImageBlock 700 0 52 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:32.819	Information	[Instrument Response] 8 ImageBlock 800 0 52 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:32.819	Information	Transferring image block at (0,52) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:56:32.850	Information	Transferring image block at (0,52) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:56:40.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/14/23 12:56:40.562	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:56:40.562	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:56:40.562	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:56:41.515	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:56:42.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      275 mA 	GMT -5:00
12/14/23 12:56:46.109	Information	[Instrument Response] 8 ImageBlock 700 0 65 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:46.109	Information	[Instrument Response] 8 ImageBlock 800 0 65 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:46.109	Information	Transferring image block at (0,65) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:56:46.140	Information	Transferring image block at (0,65) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:56:46.531	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 12:56:46.594	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:56:47.547	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:56:48.547	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:56:51.562	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:56:52.561	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:56:53.561	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:56:53.624	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         102 V  	GMT -5:00
12/14/23 12:56:53.624	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:56:53.624	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:56:54.561	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:56:56.514	Information	[Instrument Response] 8 Progress 30% 3 minutes, 30 seconds	GMT -5:00
12/14/23 12:56:58.585	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:56:58.975	Information	[Instrument Response] 8 ImageBlock 700 0 78 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:58.975	Information	[Instrument Response] 8 ImageBlock 800 0 78 337 13 337 302 8762	GMT -5:00
12/14/23 12:56:58.975	Information	Transferring image block at (0,78) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:56:59.006	Information	Transferring image block at (0,78) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:56:59.585	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:57:03.605	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:57:04.605	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:57:06.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/14/23 12:57:06.667	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:57:06.667	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 12:57:06.667	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:57:07.620	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:57:12.275	Information	[Instrument Response] 8 ImageBlock 700 0 91 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:12.275	Information	[Instrument Response] 8 ImageBlock 800 0 91 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:12.275	Information	Transferring image block at (0,91) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:57:12.322	Information	Transferring image block at (0,91) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:57:15.649	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:57:16.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:57:19.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:57:19.733	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         130 V  	GMT -5:00
12/14/23 12:57:19.733	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:57:19.733	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:57:19.733	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:57:20.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 12:57:20.733	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:57:21.685	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:57:25.138	Information	[Instrument Response] 8 ImageBlock 700 0 104 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:25.138	Information	[Instrument Response] 8 ImageBlock 800 0 104 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:25.138	Information	Transferring image block at (0,104) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:57:25.169	Information	Transferring image block at (0,104) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:57:26.435	Information	[Instrument Response] 8 Progress 40% 3 minutes	GMT -5:00
12/14/23 12:57:32.732	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/14/23 12:57:32.779	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         168 V  	GMT -5:00
12/14/23 12:57:32.779	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:57:32.779	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:57:32.779	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:57:32.779	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:57:33.732	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/14/23 12:57:33.779	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:57:33.779	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:57:33.779	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:57:34.732	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:57:38.434	Information	[Instrument Response] 8 ImageBlock 700 0 117 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:38.434	Information	[Instrument Response] 8 ImageBlock 800 0 117 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:38.434	Information	Transferring image block at (0,117) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:57:38.465	Information	Transferring image block at (0,117) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:57:46.776	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:57:46.838	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          99 V  	GMT -5:00
12/14/23 12:57:46.838	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:57:46.838	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:57:47.792	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:57:51.311	Information	[Instrument Response] 8 ImageBlock 700 0 130 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:51.311	Information	[Instrument Response] 8 ImageBlock 800 0 130 337 13 337 302 8762	GMT -5:00
12/14/23 12:57:51.311	Information	Transferring image block at (0,130) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:57:51.342	Information	Transferring image block at (0,130) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:57:56.366	Information	[Instrument Response] 8 Progress 50% 2 minutes, 30 seconds	GMT -5:00
12/14/23 12:57:59.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/14/23 12:57:59.902	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:57:59.902	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:57:59.902	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:58:00.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:58:04.593	Information	[Instrument Response] 8 ImageBlock 700 0 143 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:04.593	Information	[Instrument Response] 8 ImageBlock 800 0 143 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:04.608	Information	Transferring image block at (0,143) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:58:04.624	Information	Transferring image block at (0,143) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:58:05.874	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:58:06.873	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:58:10.885	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:58:11.884	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:58:12.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:58:12.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         125 V  	GMT -5:00
12/14/23 12:58:12.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:58:12.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:58:13.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 12:58:14.903	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:58:17.466	Information	[Instrument Response] 8 ImageBlock 700 0 156 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:17.466	Information	[Instrument Response] 8 ImageBlock 800 0 156 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:17.481	Information	Transferring image block at (0,156) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:58:17.497	Information	Transferring image block at (0,156) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:58:17.919	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:58:18.919	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:58:25.879	Information	[Instrument Response] 8 Progress 60% 2 minutes	GMT -5:00
12/14/23 12:58:25.945	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         145 V  	GMT -5:00
12/14/23 12:58:26.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/14/23 12:58:26.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:58:26.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:58:26.007	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:58:26.944	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/14/23 12:58:27.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 12:58:27.007	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:58:27.960	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:58:29.966	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:58:30.763	Information	[Instrument Response] 8 ImageBlock 700 0 169 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:30.763	Information	[Instrument Response] 8 ImageBlock 800 0 169 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:30.763	Information	Transferring image block at (0,169) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:58:30.794	Information	Transferring image block at (0,169) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:58:30.966	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:58:36.237	Information	[Instrument Response] 8 Progress 64% 1 minute, 50 seconds	GMT -5:00
12/14/23 12:58:40.002	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:58:40.064	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:58:40.064	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:58:40.064	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:58:43.635	Information	[Instrument Response] 8 ImageBlock 700 0 182 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:43.635	Information	[Instrument Response] 8 ImageBlock 800 0 182 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:43.635	Information	Transferring image block at (0,182) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:58:43.666	Information	Transferring image block at (0,182) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:58:46.597	Information	[Instrument Response] 8 Progress 67% 1 minute, 40 seconds	GMT -5:00
12/14/23 12:58:53.063	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         107 V  	GMT -5:00
12/14/23 12:58:53.110	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:58:53.110	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 12:58:53.110	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:58:54.063	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:58:56.541	Information	[Instrument Response] 8 Progress 70% 1 minute, 30 seconds	GMT -5:00
12/14/23 12:58:56.931	Information	[Instrument Response] 8 ImageBlock 700 0 195 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:56.931	Information	[Instrument Response] 8 ImageBlock 800 0 195 337 13 337 302 8762	GMT -5:00
12/14/23 12:58:56.931	Information	Transferring image block at (0,195) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:58:56.962	Information	Transferring image block at (0,195) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:59:05.746	Information	[Instrument Response] 8 Progress 74% 1 minute, 20 seconds	GMT -5:00
12/14/23 12:59:06.105	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:59:06.168	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/14/23 12:59:06.168	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:59:06.168	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:59:07.121	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:59:09.796	Information	[Instrument Response] 8 ImageBlock 700 0 208 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:09.796	Information	[Instrument Response] 8 ImageBlock 800 0 208 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:09.812	Information	Transferring image block at (0,208) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:59:09.843	Information	Transferring image block at (0,208) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:59:16.538	Information	[Instrument Response] 8 Progress 77% 1 minute, 10 seconds	GMT -5:00
12/14/23 12:59:19.178	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         139 V  	GMT -5:00
12/14/23 12:59:19.225	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:59:19.225	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:59:19.225	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:59:19.225	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:59:20.178	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/14/23 12:59:20.241	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:59:21.178	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:59:23.100	Information	[Instrument Response] 8 ImageBlock 700 0 221 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:23.100	Information	[Instrument Response] 8 ImageBlock 800 0 221 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:23.100	Information	Transferring image block at (0,221) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:59:23.131	Information	Transferring image block at (0,221) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:59:25.193	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:59:25.256	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:59:26.209	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:59:26.256	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:59:26.475	Information	[Instrument Response] 8 Progress 80% 1 minute	GMT -5:00
12/14/23 12:59:30.211	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 12:59:31.226	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:59:33.225	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:59:33.288	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 12:59:33.288	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:59:33.288	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 12:59:34.241	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 12:59:35.975	Information	[Instrument Response] 8 ImageBlock 700 0 234 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:35.975	Information	[Instrument Response] 8 ImageBlock 800 0 234 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:35.975	Information	Transferring image block at (0,234) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:59:36.022	Information	Transferring image block at (0,234) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:59:36.428	Information	[Instrument Response] 8 Progress 83% 50 seconds	GMT -5:00
12/14/23 12:59:37.240	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 12:59:37.303	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:59:38.256	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 12:59:38.302	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:59:42.271	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 12:59:42.318	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 12:59:43.273	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:59:43.320	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 12:59:45.617	Information	[Instrument Response] 8 Progress 87% 40 seconds	GMT -5:00
12/14/23 12:59:46.288	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/14/23 12:59:46.335	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:59:46.335	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 12:59:46.335	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 12:59:47.288	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 12:59:49.264	Information	[Instrument Response] 8 ImageBlock 700 0 247 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:49.264	Information	[Instrument Response] 8 ImageBlock 800 0 247 337 13 337 302 8762	GMT -5:00
12/14/23 12:59:49.264	Information	Transferring image block at (0,247) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 12:59:49.279	Information	Transferring image block at (0,247) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 12:59:49.342	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 12:59:50.295	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:59:54.325	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 12:59:55.325	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 12:59:56.408	Information	[Instrument Response] 8 Progress 91% 30 seconds	GMT -5:00
12/14/23 12:59:59.330	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 12:59:59.393	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         118 V  	GMT -5:00
12/14/23 12:59:59.393	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 12:59:59.393	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:00:00.345	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:00:02.130	Information	[Instrument Response] 8 ImageBlock 700 0 260 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:02.130	Information	[Instrument Response] 8 ImageBlock 800 0 260 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:02.130	Information	Transferring image block at (0,260) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 13:00:02.161	Information	Transferring image block at (0,260) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 13:00:06.345	Information	[Instrument Response] 8 Progress 94% 20 seconds	GMT -5:00
12/14/23 13:00:12.385	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         135 V  	GMT -5:00
12/14/23 13:00:12.448	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:00:12.448	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:00:12.448	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:00:12.448	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:00:13.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/14/23 13:00:13.447	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:00:14.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:00:15.416	Information	[Instrument Response] 8 ImageBlock 700 0 273 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:15.416	Information	[Instrument Response] 8 ImageBlock 800 0 273 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:15.432	Information	Transferring image block at (0,273) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 13:00:15.447	Information	Transferring image block at (0,273) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 13:00:16.291	Information	[Instrument Response] 8 Progress 97% 10 seconds	GMT -5:00
12/14/23 13:00:22.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:00:22.483	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/14/23 13:00:22.483	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:00:22.483	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:00:23.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:00:24.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:00:24.945	Information	[Instrument Response] 8 ImageBlock 700 0 286 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:24.945	Information	[Instrument Response] 8 ImageBlock 800 0 286 337 13 337 302 8762	GMT -5:00
12/14/23 13:00:24.960	Information	Transferring image block at (0,286) with size 337x13 to the 700 channel.	GMT -5:00
12/14/23 13:00:24.976	Information	Transferring image block at (0,286) with size 337x13 to the 800 channel.	GMT -5:00
12/14/23 13:00:26.240	Information	[Instrument Response] 8 Progress 100% 0 seconds	GMT -5:00
12/14/23 13:00:26.255	Information	[Instrument Response] 8 ImageBlock 700 0 299 337 3 337 302 2022	GMT -5:00
12/14/23 13:00:26.255	Information	[Instrument Response] 8 ImageBlock 800 0 299 337 3 337 302 2022	GMT -5:00
12/14/23 13:00:26.255	Information	[Instrument Response] 8 Status 0 COMPLETE SCAN 300.582 seconds	GMT -5:00
12/14/23 13:00:26.255	Information	Transferring image block at (0,299) with size 337x3 to the 700 channel.	GMT -5:00
12/14/23 13:00:26.255	Information	[Instrument Response] 8 Status 0 Averages (sec): setup=2.00005 line=0.41818 => total=255	GMT -5:00
12/14/23 13:00:26.287	Information	Transferring image block at (0,299) with size 337x3 to the 800 channel.	GMT -5:00
12/14/23 13:00:26.302	Information	[Instrument Response] 8 Status 0 Comparisons: setup=1.00997 line=0.952512	GMT -5:00
12/14/23 13:00:26.349	Information	[Instrument Response] 8 Status 0 Finalize motors	GMT -5:00
12/14/23 13:00:26.349	Information	[Instrument Response] 8 Progress 100% Scan complete	GMT -5:00
12/14/23 13:00:26.365	Information	[Instrument Response] 8 Result 0 OK	GMT -5:00
12/14/23 13:00:26.365	Information	Acquisition 0000002_01 finished with channels: 700 800.	GMT -5:00
12/14/23 13:00:26.724	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/14/23 13:00:26.724	Information	Scan completed successfully	GMT -5:00
12/14/23 13:00:26.833	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         139 V  	GMT -5:00
12/14/23 13:00:26.849	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          87 V  	GMT -5:00
12/14/23 13:00:26.849	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 13:00:26.849	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 13:00:26.849	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 13:00:27.455	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          81 V  	GMT -5:00
12/14/23 13:00:27.501	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          51 V  	GMT -5:00
12/14/23 13:00:28.456	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          50 V  	GMT -5:00
12/14/23 13:00:28.501	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          31 V  	GMT -5:00
12/14/23 13:00:29.461	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          31 V  	GMT -5:00
12/14/23 13:00:29.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          20 V  	GMT -5:00
12/14/23 13:00:30.462	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          21 V  	GMT -5:00
12/14/23 13:00:30.509	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/14/23 13:00:31.463	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          14 V  	GMT -5:00
12/14/23 13:00:31.526	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/14/23 13:00:32.464	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          10 V  	GMT -5:00
12/14/23 13:00:32.527	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/14/23 13:00:33.465	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/14/23 13:00:34.482	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/14/23 13:00:34.528	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:00:36.485	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:00:36.531	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      244 mA 	GMT -5:00
12/14/23 13:00:39.489	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:00:40.496	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:02:27.919	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/14/23 13:05:37.804	Information	Starting acquisition...	GMT -5:00
12/14/23 13:05:38.650	Information	[Instrument Request]  9 ScanAuto 169um medium 0.0 0.0 5.7 0.0 5.1 700 800 	GMT -5:00
12/14/23 13:05:43.556	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/14/23 13:05:54.871	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/14/23 13:06:02.555	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/14/23 13:06:02.602	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/14/23 13:06:02.649	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/14/23 13:06:02.664	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/14/23 13:06:02.664	Information	[Instrument Request]  10 ScanAutoEstimate 169um medium 0.0 0.0 5.7 0.0 5.1 700 800 	GMT -5:00
12/14/23 13:06:05.619	Information	[Instrument Response] 10 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0 5.7 0 5.1 700 800	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 X mapping: [x0=0,x1=5.7] => [left=-60,right=2644] w=337	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Y mapping: [y0=0,y1=5.1] => [top=34,bottom=2450] h=302	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Seconds/line=0.375775	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 y_scan_pixels=302, passes=2 => lines_to_scan=604	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 linesPerBlock=27 flushLinesPerBlock=13	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Blocks=23	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Status 0 Estimates (sec): setup=2.02 line=0.398321 => total=243	GMT -5:00
12/14/23 13:06:05.634	Information	[Instrument Response] 10 Progress 0% 5 minutes	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 10 Result 0 OK	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/14/23 13:06:05.665	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/14/23 13:06:05.681	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/14/23 13:06:13.052	Information	[Instrument Request]  11 ScanAutoEstimate 169um medium 0.0 0.1 8.5 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:06:13.067	Information	[Instrument Response] 11 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 8.5 0 4.9 700 800	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 X mapping: [x0=0.1,x1=8.5] => [left=-13,right=3971] w=497	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Seconds/line=0.510775	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Blocks=30	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Status 0 Estimates (sec): setup=2.02 line=0.541421 => total=317	GMT -5:00
12/14/23 13:06:13.115	Information	[Instrument Response] 11 Progress 0% 7 minutes	GMT -5:00
12/14/23 13:06:13.130	Information	[Instrument Response] 11 Result 0 OK	GMT -5:00
12/14/23 13:06:16.090	Information	Starting acquisition...	GMT -5:00
12/14/23 13:06:16.886	Information	[Instrument Request]  12 ScanAuto 169um medium 0.0 0.1 8.5 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 1 Executing command: ScanAuto 169um medium 0 0.1 8.5 0 4.9 700 800	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 X mapping: [x0=0.1,x1=8.5] => [left=-13,right=3971] w=497	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Seconds/line=0.510775	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Blocks=30	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Status 0 Estimates (sec): setup=2.02 line=0.541421 => total=317	GMT -5:00
12/14/23 13:06:16.964	Information	[Instrument Response] 12 Progress 0% 7 minutes	GMT -5:00
12/14/23 13:06:16.980	Information	[Instrument Response] 12 Progress 0% Initializing	GMT -5:00
12/14/23 13:06:16.996	Information	[Instrument Response] 12 Status 0 Preparation complete	GMT -5:00
12/14/23 13:06:19.905	Information	[Instrument Response] 12 Status 0 Motors initialized	GMT -5:00
12/14/23 13:06:19.967	Information	[Instrument Response] 12 Status 0 X-Motor configured	GMT -5:00
12/14/23 13:06:19.967	Information	[Instrument Response] 12 Status 0 DSP configured	GMT -5:00
12/14/23 13:06:19.967	Information	[Instrument Response] 12 Status 0 Digital / DSP configured	GMT -5:00
12/14/23 13:06:19.967	Information	[Instrument Response] 12 Status 0 Cooling	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 Create MergeImageChannel: 700 20 0 0 497 10 497 290 xofs=1 yofs=0	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 Create MergeImageChannel: 800 20 0 0 497 10 497 290 xofs=0 yofs=0	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 start=34 scan_lines=10 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 start=34 scan_lines=20 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/14/23 13:06:20.014	Information	[Instrument Response] 12 Status 0 Move motors to initial positions	GMT -5:00
12/14/23 13:06:20.686	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/14/23 13:06:20.749	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:06:23.704	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:06:23.954	Information	[Instrument Response] 12 Progress 0% 7 minutes	GMT -5:00
12/14/23 13:06:24.704	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:06:24.767	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:06:24.767	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 13:06:24.767	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:06:24.767	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:06:28.716	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:06:29.716	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:06:30.732	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/14/23 13:06:31.731	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:06:31.794	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         129 V  	GMT -5:00
12/14/23 13:06:31.794	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:06:31.794	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:06:32.732	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:06:33.733	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:06:38.963	Information	[Instrument Response] 12 ImageBlock 700 0 0 497 10 497 290 9940	GMT -5:00
12/14/23 13:06:38.963	Information	[Instrument Response] 12 ImageBlock 800 0 0 497 10 497 290 9940	GMT -5:00
12/14/23 13:06:38.963	Information	Transferring image block at (0,0) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:06:38.994	Information	Transferring image block at (0,0) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:06:44.531	Information	[Instrument Response] 12 Progress 5% 6 minutes	GMT -5:00
12/14/23 13:06:44.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         128 V  	GMT -5:00
12/14/23 13:06:44.844	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:06:44.844	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:06:44.844	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:06:45.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:06:46.796	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:06:49.811	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:06:50.815	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:06:52.006	Information	[Instrument Response] 12 ImageBlock 700 0 10 497 10 497 290 9940	GMT -5:00
12/14/23 13:06:52.006	Information	[Instrument Response] 12 ImageBlock 800 0 10 497 10 497 290 9940	GMT -5:00
12/14/23 13:06:52.021	Information	Transferring image block at (0,10) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:06:52.037	Information	Transferring image block at (0,10) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:06:57.837	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:06:57.899	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/14/23 13:06:57.899	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:06:57.899	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:06:58.837	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:06:59.840	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:07:05.047	Information	[Instrument Response] 12 ImageBlock 700 0 20 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:05.047	Information	[Instrument Response] 12 ImageBlock 800 0 20 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:05.062	Information	Transferring image block at (0,20) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:07:05.094	Information	Transferring image block at (0,20) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:07:10.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         126 V  	GMT -5:00
12/14/23 13:07:10.947	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:07:10.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:07:10.947	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:07:11.907	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:07:12.907	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:07:15.908	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:07:16.924	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:07:18.095	Information	[Instrument Response] 12 ImageBlock 700 0 30 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:18.095	Information	[Instrument Response] 12 ImageBlock 800 0 30 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:18.095	Information	Transferring image block at (0,30) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:07:18.142	Information	Transferring image block at (0,30) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:07:23.945	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:07:24.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/14/23 13:07:24.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:07:24.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:07:24.954	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:07:25.953	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:07:31.140	Information	[Instrument Response] 12 ImageBlock 700 0 40 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:31.140	Information	[Instrument Response] 12 ImageBlock 800 0 40 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:31.140	Information	Transferring image block at (0,40) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:07:31.171	Information	Transferring image block at (0,40) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:07:36.992	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         125 V  	GMT -5:00
12/14/23 13:07:37.055	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:07:37.055	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:07:37.055	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:07:38.008	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:07:44.188	Information	[Instrument Response] 12 ImageBlock 700 0 50 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:44.188	Information	[Instrument Response] 12 ImageBlock 800 0 50 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:44.188	Information	Transferring image block at (0,50) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:07:44.219	Information	Transferring image block at (0,50) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:07:46.437	Information	[Instrument Response] 12 Progress 21% 5 minutes	GMT -5:00
12/14/23 13:07:50.062	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:07:50.109	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         125 V  	GMT -5:00
12/14/23 13:07:50.109	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:07:50.109	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:07:51.065	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:07:52.065	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:07:57.239	Information	[Instrument Response] 12 ImageBlock 700 0 60 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:57.239	Information	[Instrument Response] 12 ImageBlock 800 0 60 497 10 497 290 9940	GMT -5:00
12/14/23 13:07:57.239	Information	Transferring image block at (0,60) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:07:57.270	Information	Transferring image block at (0,60) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:08:03.108	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         124 V  	GMT -5:00
12/14/23 13:08:03.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:08:03.170	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 13:08:03.170	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:08:04.121	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:08:10.282	Information	[Instrument Response] 12 ImageBlock 700 0 70 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:10.282	Information	[Instrument Response] 12 ImageBlock 800 0 70 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:10.282	Information	Transferring image block at (0,70) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:08:10.313	Information	Transferring image block at (0,70) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:08:14.157	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:08:15.156	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:08:15.844	Information	[Instrument Response] 12 Progress 29% 4 minutes, 30 seconds	GMT -5:00
12/14/23 13:08:16.156	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:08:16.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/14/23 13:08:16.219	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:08:16.219	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:08:17.172	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:08:18.171	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:08:23.331	Information	[Instrument Response] 12 ImageBlock 700 0 80 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:23.331	Information	[Instrument Response] 12 ImageBlock 800 0 80 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:23.331	Information	Transferring image block at (0,80) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:08:23.362	Information	Transferring image block at (0,80) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:08:27.204	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:08:28.204	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:08:29.220	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         123 V  	GMT -5:00
12/14/23 13:08:29.266	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:08:29.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:08:29.266	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:08:30.220	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:08:36.376	Information	[Instrument Response] 12 ImageBlock 700 0 90 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:36.376	Information	[Instrument Response] 12 ImageBlock 800 0 90 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:36.376	Information	Transferring image block at (0,90) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:08:36.407	Information	Transferring image block at (0,90) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:08:40.267	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:08:41.267	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:08:42.267	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:08:42.329	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/14/23 13:08:42.329	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:08:42.329	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:08:43.282	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:08:44.282	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:08:46.703	Information	[Instrument Response] 12 Progress 37% 4 minutes	GMT -5:00
12/14/23 13:08:49.428	Information	[Instrument Response] 12 ImageBlock 700 0 100 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:49.428	Information	[Instrument Response] 12 ImageBlock 800 0 100 497 10 497 290 9940	GMT -5:00
12/14/23 13:08:49.428	Information	Transferring image block at (0,100) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:08:49.459	Information	Transferring image block at (0,100) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:08:53.321	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:08:54.321	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:08:55.321	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:08:55.383	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:08:55.383	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:08:55.383	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:08:56.321	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:09:02.474	Information	[Instrument Response] 12 ImageBlock 700 0 110 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:02.474	Information	[Instrument Response] 12 ImageBlock 800 0 110 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:02.474	Information	Transferring image block at (0,110) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:09:02.505	Information	Transferring image block at (0,110) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:09:06.364	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:09:07.373	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:09:08.380	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:09:08.426	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         123 V  	GMT -5:00
12/14/23 13:09:08.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:09:08.426	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:09:09.379	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:09:15.522	Information	[Instrument Response] 12 ImageBlock 700 0 120 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:15.522	Information	[Instrument Response] 12 ImageBlock 800 0 120 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:15.522	Information	Transferring image block at (0,120) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:09:15.553	Information	Transferring image block at (0,120) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:09:16.662	Information	[Instrument Response] 12 Progress 45% 3 minutes, 30 seconds	GMT -5:00
12/14/23 13:09:19.417	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:09:20.417	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:09:21.432	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:09:21.479	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:09:21.479	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:09:21.479	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:09:22.432	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:09:28.572	Information	[Instrument Response] 12 ImageBlock 700 0 130 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:28.572	Information	[Instrument Response] 12 ImageBlock 800 0 130 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:28.572	Information	Transferring image block at (0,130) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:09:28.603	Information	Transferring image block at (0,130) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:09:32.462	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:09:33.477	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:09:34.477	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:09:34.540	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:09:34.540	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:09:34.540	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:09:35.477	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:09:41.618	Information	[Instrument Response] 12 ImageBlock 700 0 140 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:41.618	Information	[Instrument Response] 12 ImageBlock 800 0 140 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:41.618	Information	Transferring image block at (0,140) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:09:41.649	Information	Transferring image block at (0,140) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:09:46.632	Information	[Instrument Response] 12 Progress 53% 3 minutes	GMT -5:00
12/14/23 13:09:47.523	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:09:47.585	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:09:47.585	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:09:47.585	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:09:48.538	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:09:54.666	Information	[Instrument Response] 12 ImageBlock 700 0 150 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:54.666	Information	[Instrument Response] 12 ImageBlock 800 0 150 497 10 497 290 9940	GMT -5:00
12/14/23 13:09:54.666	Information	Transferring image block at (0,150) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:09:54.697	Information	Transferring image block at (0,150) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:10:00.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:10:00.645	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:10:00.645	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:10:00.645	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:10:01.591	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:10:07.715	Information	[Instrument Response] 12 ImageBlock 700 0 160 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:07.715	Information	[Instrument Response] 12 ImageBlock 800 0 160 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:07.715	Information	Transferring image block at (0,160) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:10:07.746	Information	Transferring image block at (0,160) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:10:13.636	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/14/23 13:10:13.698	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:10:13.698	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:10:13.698	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:10:14.651	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:10:16.380	Information	[Instrument Response] 12 Progress 61% 2 minutes, 30 seconds	GMT -5:00
12/14/23 13:10:20.762	Information	[Instrument Response] 12 ImageBlock 700 0 170 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:20.762	Information	[Instrument Response] 12 ImageBlock 800 0 170 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:20.777	Information	Transferring image block at (0,170) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:10:20.808	Information	Transferring image block at (0,170) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:10:26.689	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:10:26.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         121 V  	GMT -5:00
12/14/23 13:10:26.751	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:10:26.751	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:10:27.704	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:10:32.719	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:10:33.718	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:10:33.812	Information	[Instrument Response] 12 ImageBlock 700 0 180 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:33.812	Information	[Instrument Response] 12 ImageBlock 800 0 180 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:33.812	Information	Transferring image block at (0,180) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:10:33.843	Information	Transferring image block at (0,180) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:10:39.750	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/14/23 13:10:39.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:10:39.797	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:10:39.797	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:10:40.743	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/14/23 13:10:45.773	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:10:46.351	Information	[Instrument Response] 12 Progress 69% 2 minutes	GMT -5:00
12/14/23 13:10:46.773	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:10:46.851	Information	[Instrument Response] 12 ImageBlock 700 0 190 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:46.867	Information	[Instrument Response] 12 ImageBlock 800 0 190 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:46.867	Information	Transferring image block at (0,190) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:10:46.882	Information	Transferring image block at (0,190) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:10:52.808	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:10:52.855	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         120 V  	GMT -5:00
12/14/23 13:10:52.855	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:10:52.855	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:10:53.810	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:10:56.628	Information	[Instrument Response] 12 Progress 71% 1 minute, 50 seconds	GMT -5:00
12/14/23 13:10:58.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:10:58.878	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:10:59.831	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:10:59.878	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:10:59.909	Information	[Instrument Response] 12 ImageBlock 700 0 200 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:59.909	Information	[Instrument Response] 12 ImageBlock 800 0 200 497 10 497 290 9940	GMT -5:00
12/14/23 13:10:59.909	Information	Transferring image block at (0,200) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:10:59.940	Information	Transferring image block at (0,200) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:11:05.466	Information	[Instrument Response] 12 Progress 74% 1 minute, 40 seconds	GMT -5:00
12/14/23 13:11:05.856	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         119 V  	GMT -5:00
12/14/23 13:11:05.903	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:11:05.903	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:11:05.903	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:11:06.856	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:11:11.873	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:11:12.881	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:11:12.944	Information	[Instrument Response] 12 ImageBlock 700 0 210 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:12.944	Information	[Instrument Response] 12 ImageBlock 800 0 210 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:12.959	Information	Transferring image block at (0,210) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:11:12.991	Information	Transferring image block at (0,210) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:11:16.847	Information	[Instrument Response] 12 Progress 77% 1 minute, 30 seconds	GMT -5:00
12/14/23 13:11:18.903	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:11:18.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/14/23 13:11:18.965	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:11:18.965	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:11:19.903	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:11:25.984	Information	[Instrument Response] 12 ImageBlock 700 0 220 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:25.984	Information	[Instrument Response] 12 ImageBlock 800 0 220 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:26.000	Information	Transferring image block at (0,220) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:11:26.031	Information	Transferring image block at (0,220) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:11:26.578	Information	[Instrument Response] 12 Progress 79% 1 minute, 20 seconds	GMT -5:00
12/14/23 13:11:31.960	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         118 V  	GMT -5:00
12/14/23 13:11:32.006	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:11:32.006	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:11:32.006	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:11:32.959	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:11:36.320	Information	[Instrument Response] 12 Progress 82% 1 minute, 10 seconds	GMT -5:00
12/14/23 13:11:39.038	Information	[Instrument Response] 12 ImageBlock 700 0 230 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:39.038	Information	[Instrument Response] 12 ImageBlock 800 0 230 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:39.038	Information	Transferring image block at (0,230) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:11:39.069	Information	Transferring image block at (0,230) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:11:44.597	Information	[Instrument Response] 12 Progress 84% 1 minute	GMT -5:00
12/14/23 13:11:45.019	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:11:45.066	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         118 V  	GMT -5:00
12/14/23 13:11:45.066	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:11:45.066	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:11:46.019	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:11:52.082	Information	[Instrument Response] 12 ImageBlock 700 0 240 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:52.082	Information	[Instrument Response] 12 ImageBlock 800 0 240 497 10 497 290 9940	GMT -5:00
12/14/23 13:11:52.097	Information	Transferring image block at (0,240) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:11:52.113	Information	Transferring image block at (0,240) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:11:56.550	Information	[Instrument Response] 12 Progress 88% 50 seconds	GMT -5:00
12/14/23 13:11:58.066	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         117 V  	GMT -5:00
12/14/23 13:11:58.129	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:11:58.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:11:58.129	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:11:59.066	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:12:05.136	Information	[Instrument Response] 12 ImageBlock 700 0 250 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:05.136	Information	[Instrument Response] 12 ImageBlock 800 0 250 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:05.136	Information	Transferring image block at (0,250) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:12:05.167	Information	Transferring image block at (0,250) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:12:06.829	Information	[Instrument Response] 12 Progress 90% 40 seconds	GMT -5:00
12/14/23 13:12:10.120	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/14/23 13:12:11.120	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:12:11.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/14/23 13:12:11.183	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         117 V  	GMT -5:00
12/14/23 13:12:11.183	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:12:11.183	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:12:12.120	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:12:16.557	Information	[Instrument Response] 12 Progress 93% 30 seconds	GMT -5:00
12/14/23 13:12:18.168	Information	[Instrument Response] 12 ImageBlock 700 0 260 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:18.183	Information	[Instrument Response] 12 ImageBlock 800 0 260 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:18.183	Information	Transferring image block at (0,260) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:12:18.215	Information	Transferring image block at (0,260) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:12:23.169	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:12:23.231	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:12:24.176	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         116 V  	GMT -5:00
12/14/23 13:12:24.223	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:12:24.223	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 13:12:24.223	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:12:24.223	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:12:24.223	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:12:25.176	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:12:26.832	Information	[Instrument Response] 12 Progress 95% 20 seconds	GMT -5:00
12/14/23 13:12:31.220	Information	[Instrument Response] 12 ImageBlock 700 0 270 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:31.220	Information	[Instrument Response] 12 ImageBlock 800 0 270 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:31.220	Information	Transferring image block at (0,270) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:12:31.251	Information	Transferring image block at (0,270) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:12:36.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/14/23 13:12:36.282	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:12:36.782	Information	[Instrument Response] 12 Progress 98% 10 seconds	GMT -5:00
12/14/23 13:12:37.219	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:12:37.281	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/14/23 13:12:37.281	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         117 V  	GMT -5:00
12/14/23 13:12:37.281	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:12:37.281	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:12:37.281	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:12:38.234	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:12:44.256	Information	[Instrument Response] 12 ImageBlock 700 0 280 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:44.256	Information	[Instrument Response] 12 ImageBlock 800 0 280 497 10 497 290 9940	GMT -5:00
12/14/23 13:12:44.272	Information	Transferring image block at (0,280) with size 497x10 to the 700 channel.	GMT -5:00
12/14/23 13:12:44.303	Information	Transferring image block at (0,280) with size 497x10 to the 800 channel.	GMT -5:00
12/14/23 13:12:44.350	Information	[Instrument Response] 12 Progress 100% 0 seconds	GMT -5:00
12/14/23 13:12:44.365	Information	[Instrument Response] 12 Status 0 COMPLETE SCAN 380.343 seconds	GMT -5:00
12/14/23 13:12:44.365	Information	[Instrument Response] 12 Status 0 Averages (sec): setup=2.00005 line=0.552313 => total=323	GMT -5:00
12/14/23 13:12:44.365	Information	[Instrument Response] 12 Status 0 Comparisons: setup=1.00997 line=0.98028	GMT -5:00
12/14/23 13:12:44.365	Information	[Instrument Response] 12 Status 0 Finalize motors	GMT -5:00
12/14/23 13:12:44.365	Information	[Instrument Response] 12 Progress 100% Scan complete	GMT -5:00
12/14/23 13:12:44.413	Information	[Instrument Response] 12 Result 0 OK	GMT -5:00
12/14/23 13:12:44.413	Information	Acquisition 0000003_01 finished with channels: 700 800.	GMT -5:00
12/14/23 13:12:44.725	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/14/23 13:12:44.725	Information	Scan completed successfully	GMT -5:00
12/14/23 13:12:45.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/14/23 13:12:45.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          57 V  	GMT -5:00
12/14/23 13:12:45.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 13:12:45.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 13:12:45.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 13:12:46.262	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          56 V  	GMT -5:00
12/14/23 13:12:46.312	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          35 V  	GMT -5:00
12/14/23 13:12:47.262	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          35 V  	GMT -5:00
12/14/23 13:12:47.312	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          22 V  	GMT -5:00
12/14/23 13:12:48.267	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          23 V  	GMT -5:00
12/14/23 13:12:48.330	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          14 V  	GMT -5:00
12/14/23 13:12:49.269	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          15 V  	GMT -5:00
12/14/23 13:12:49.332	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          10 V  	GMT -5:00
12/14/23 13:12:50.271	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          11 V  	GMT -5:00
12/14/23 13:12:50.333	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           7 V  	GMT -5:00
12/14/23 13:12:51.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/14/23 13:12:51.335	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           4 V  	GMT -5:00
12/14/23 13:12:52.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/14/23 13:12:53.291	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:12:54.361	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:12:54.393	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/14/23 13:12:57.299	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:12:59.309	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:19:30.817	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    736 mV 	GMT -5:00
12/14/23 13:23:19.393	Information	Starting acquisition...	GMT -5:00
12/14/23 13:23:20.189	Information	[Instrument Request]  13 ScanAuto 169um medium 0.0 0.1 8.5 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:23:25.063	Error      	The instrument connection was closed. If a connection is expected, check the cable connection and confirm the instrument is powered on.  To reconnect, click the Image Studio Application Button, hover over Instrument, and click Connect.	GMT -5:00
12/14/23 13:23:39.868	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/14/23 13:23:47.573	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/14/23 13:23:47.620	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/14/23 13:23:47.667	Information	[Instrument Request]  14 ScanAutoEstimate 169um medium 0.0 0.1 8.5 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 8.5 0 4.9 700 800	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 X mapping: [x0=0.1,x1=8.5] => [left=-13,right=3971] w=497	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Seconds/line=0.510775	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 linesPerBlock=20 flushLinesPerBlock=10	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Blocks=30	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Status 0 Estimates (sec): setup=2.02 line=0.541421 => total=317	GMT -5:00
12/14/23 13:23:50.651	Information	[Instrument Response] 14 Progress 0% 7 minutes	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 14 Result 0 OK	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    666 mV 	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/14/23 13:23:50.682	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/14/23 13:23:50.713	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/14/23 13:23:56.761	Information	[Instrument Request]  15 ScanAutoEstimate 169um medium 0.0 0.1 4.2 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:23:56.777	Information	[Instrument Response] 15 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 4.2 0 4.9 700 800	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 X mapping: [x0=0.1,x1=4.2] => [left=-13,right=1939] w=243	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Seconds/line=0.295775	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 linesPerBlock=34 flushLinesPerBlock=17	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Blocks=18	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Status 0 Estimates (sec): setup=2.02 line=0.313521 => total=184	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Progress 0% 4 minutes	GMT -5:00
12/14/23 13:23:56.808	Information	[Instrument Response] 15 Result 0 OK	GMT -5:00
12/14/23 13:23:58.401	Information	Starting acquisition...	GMT -5:00
12/14/23 13:23:59.261	Information	[Instrument Request]  16 ScanAuto 169um medium 0.0 0.1 4.2 0.0 4.9 700 800 	GMT -5:00
12/14/23 13:23:59.323	Information	[Instrument Response] 16 Status 1 Executing command: ScanAuto 169um medium 0 0.1 4.2 0 4.9 700 800	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 X mapping: [x0=0.1,x1=4.2] => [left=-13,right=1939] w=243	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Computed scan parameters	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Seconds/line=0.295775	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 linesPerBlock=34 flushLinesPerBlock=17	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Blocks=18	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Status 0 Estimates (sec): setup=2.02 line=0.313521 => total=184	GMT -5:00
12/14/23 13:23:59.339	Information	[Instrument Response] 16 Progress 0% 4 minutes	GMT -5:00
12/14/23 13:23:59.354	Information	[Instrument Response] 16 Progress 0% Initializing	GMT -5:00
12/14/23 13:23:59.370	Information	[Instrument Response] 16 Status 0 Preparation complete	GMT -5:00
12/14/23 13:24:02.289	Information	[Instrument Response] 16 Status 0 Motors initialized	GMT -5:00
12/14/23 13:24:02.336	Information	[Instrument Response] 16 Status 0 X-Motor configured	GMT -5:00
12/14/23 13:24:02.336	Information	[Instrument Response] 16 Status 0 DSP configured	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Digital / DSP configured	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Cooling	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Create MergeImageChannel: 700 20 0 0 243 17 243 290 xofs=1 yofs=0	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Create MergeImageChannel: 800 20 0 0 243 17 243 290 xofs=0 yofs=0	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 start=34 scan_lines=17 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 start=34 scan_lines=34 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/14/23 13:24:02.382	Information	[Instrument Response] 16 Status 0 Move motors to initial positions	GMT -5:00
12/14/23 13:24:02.695	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/14/23 13:24:02.742	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:24:03.695	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/14/23 13:24:05.699	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:24:06.167	Information	[Instrument Response] 16 Progress 0% 4 minutes	GMT -5:00
12/14/23 13:24:06.707	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:24:06.770	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:24:06.770	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/14/23 13:24:06.770	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:24:06.770	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:24:07.707	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:24:11.727	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/14/23 13:24:14.746	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:24:14.793	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          97 V  	GMT -5:00
12/14/23 13:24:14.793	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:24:14.793	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:24:20.762	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      315 mA 	GMT -5:00
12/14/23 13:24:21.605	Information	[Instrument Response] 16 ImageBlock 700 0 0 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:21.605	Information	[Instrument Response] 16 ImageBlock 800 0 0 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:21.621	Information	Transferring image block at (0,0) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:24:21.636	Information	Transferring image block at (0,0) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:24:27.790	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         118 V  	GMT -5:00
12/14/23 13:24:27.853	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/14/23 13:24:27.853	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:24:27.853	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:24:27.853	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:24:28.796	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:24:28.859	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/14/23 13:24:30.062	Information	[Instrument Response] 16 Progress 9% 3 minutes, 30 seconds	GMT -5:00
12/14/23 13:24:35.080	Information	[Instrument Response] 16 ImageBlock 700 0 17 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:35.080	Information	[Instrument Response] 16 ImageBlock 800 0 17 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:35.095	Information	Transferring image block at (0,17) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:24:35.111	Information	Transferring image block at (0,17) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:24:39.845	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:24:40.844	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:24:40.907	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/14/23 13:24:41.848	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:24:41.910	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/14/23 13:24:41.910	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:24:41.910	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:24:41.910	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:24:41.910	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:24:42.848	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:24:43.863	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:24:45.863	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:24:48.566	Information	[Instrument Response] 16 ImageBlock 700 0 34 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:48.566	Information	[Instrument Response] 16 ImageBlock 800 0 34 243 17 243 290 8262	GMT -5:00
12/14/23 13:24:48.581	Information	Transferring image block at (0,34) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:24:48.597	Information	Transferring image block at (0,34) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:24:54.916	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/14/23 13:24:54.978	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:24:54.978	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:24:54.978	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:24:55.931	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:25:00.055	Information	[Instrument Response] 16 Progress 22% 3 minutes	GMT -5:00
12/14/23 13:25:02.055	Information	[Instrument Response] 16 ImageBlock 700 0 51 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:02.055	Information	[Instrument Response] 16 ImageBlock 800 0 51 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:02.055	Information	Transferring image block at (0,51) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:25:02.086	Information	Transferring image block at (0,51) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:25:07.965	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/14/23 13:25:08.027	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         140 V  	GMT -5:00
12/14/23 13:25:08.027	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:25:08.027	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:25:08.027	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:25:08.027	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/14/23 13:25:08.980	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/14/23 13:25:09.027	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:25:09.980	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:25:15.548	Information	[Instrument Response] 16 ImageBlock 700 0 68 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:15.548	Information	[Instrument Response] 16 ImageBlock 800 0 68 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:15.548	Information	Transferring image block at (0,68) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:25:15.564	Information	Transferring image block at (0,68) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:25:20.033	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:25:22.043	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/14/23 13:25:22.090	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:25:22.090	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:25:22.090	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:25:23.043	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:25:28.066	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:25:28.113	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:25:29.035	Information	[Instrument Response] 16 ImageBlock 700 0 85 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:29.035	Information	[Instrument Response] 16 ImageBlock 800 0 85 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:29.035	Information	Transferring image block at (0,85) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:25:29.050	Information	Transferring image block at (0,85) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:25:29.113	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:25:29.113	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:25:30.097	Information	[Instrument Response] 16 Progress 36% 2 minutes, 30 seconds	GMT -5:00
12/14/23 13:25:35.104	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:25:35.150	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         128 V  	GMT -5:00
12/14/23 13:25:35.150	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:25:35.150	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:25:35.150	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:25:36.104	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:25:36.166	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:25:37.103	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:25:42.528	Information	[Instrument Response] 16 ImageBlock 700 0 102 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:42.528	Information	[Instrument Response] 16 ImageBlock 800 0 102 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:42.528	Information	Transferring image block at (0,102) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:25:42.559	Information	Transferring image block at (0,102) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:25:49.152	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:25:49.214	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:25:49.214	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:25:49.214	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:25:56.023	Information	[Instrument Response] 16 ImageBlock 700 0 119 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:56.023	Information	[Instrument Response] 16 ImageBlock 800 0 119 243 17 243 290 8262	GMT -5:00
12/14/23 13:25:56.023	Information	Transferring image block at (0,119) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:25:56.055	Information	Transferring image block at (0,119) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:26:00.114	Information	[Instrument Response] 16 Progress 49% 2 minutes	GMT -5:00
12/14/23 13:26:02.207	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:26:02.254	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/14/23 13:26:02.254	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:26:02.254	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:26:03.216	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:26:09.502	Information	[Instrument Response] 16 ImageBlock 700 0 136 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:09.517	Information	[Instrument Response] 16 ImageBlock 800 0 136 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:09.517	Information	Transferring image block at (0,136) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:26:09.549	Information	Transferring image block at (0,136) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:26:10.223	Information	[Instrument Response] 16 Progress 53% 1 minute, 50 seconds	GMT -5:00
12/14/23 13:26:11.238	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:26:11.301	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:26:12.242	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/14/23 13:26:13.258	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:26:14.257	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:26:15.273	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:26:16.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/14/23 13:26:16.320	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:26:16.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:26:16.320	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:26:17.273	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:26:20.350	Information	[Instrument Response] 16 Progress 57% 1 minute, 40 seconds	GMT -5:00
12/14/23 13:26:23.004	Information	[Instrument Response] 16 ImageBlock 700 0 153 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:23.004	Information	[Instrument Response] 16 ImageBlock 800 0 153 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:23.004	Information	Transferring image block at (0,153) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:26:23.035	Information	Transferring image block at (0,153) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:26:28.785	Information	[Instrument Response] 16 Progress 62% 1 minute, 30 seconds	GMT -5:00
12/14/23 13:26:29.320	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:26:29.383	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/14/23 13:26:29.383	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:26:29.383	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:26:30.320	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:26:36.500	Information	[Instrument Response] 16 ImageBlock 700 0 170 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:36.500	Information	[Instrument Response] 16 ImageBlock 800 0 170 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:36.500	Information	Transferring image block at (0,170) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:26:36.531	Information	Transferring image block at (0,170) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:26:40.263	Information	[Instrument Response] 16 Progress 66% 1 minute, 20 seconds	GMT -5:00
12/14/23 13:26:42.372	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         144 V  	GMT -5:00
12/14/23 13:26:42.434	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/14/23 13:26:42.434	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:26:42.434	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:26:42.434	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:26:43.385	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/14/23 13:26:43.435	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:26:43.435	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:26:44.388	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:26:49.996	Information	[Instrument Response] 16 ImageBlock 700 0 187 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:49.996	Information	[Instrument Response] 16 ImageBlock 800 0 187 243 17 243 290 8262	GMT -5:00
12/14/23 13:26:49.996	Information	Transferring image block at (0,187) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:26:50.027	Information	Transferring image block at (0,187) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:26:50.371	Information	[Instrument Response] 16 Progress 71% 1 minute, 10 seconds	GMT -5:00
12/14/23 13:26:56.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:26:56.499	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         105 V  	GMT -5:00
12/14/23 13:26:56.499	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:26:56.499	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:26:57.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:26:58.451	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:27:00.451	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:27:00.514	Information	[Instrument Response] 16 Progress 75% 1 minute	GMT -5:00
12/14/23 13:27:01.470	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:27:03.495	Information	[Instrument Response] 16 ImageBlock 700 0 204 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:03.495	Information	[Instrument Response] 16 ImageBlock 800 0 204 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:03.495	Information	Transferring image block at (0,204) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:27:03.526	Information	Transferring image block at (0,204) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:27:09.295	Information	[Instrument Response] 16 Progress 79% 50 seconds	GMT -5:00
12/14/23 13:27:09.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         132 V  	GMT -5:00
12/14/23 13:27:09.545	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:27:09.545	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:27:09.545	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:27:09.545	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/14/23 13:27:10.498	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/14/23 13:27:10.560	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:27:11.497	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:27:17.005	Information	[Instrument Response] 16 ImageBlock 700 0 221 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:17.005	Information	[Instrument Response] 16 ImageBlock 800 0 221 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:17.020	Information	Transferring image block at (0,221) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:27:17.036	Information	Transferring image block at (0,221) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:27:20.443	Information	[Instrument Response] 16 Progress 84% 40 seconds	GMT -5:00
12/14/23 13:27:23.556	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:27:23.602	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         102 V  	GMT -5:00
12/14/23 13:27:23.602	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:27:23.602	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:27:24.555	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:27:30.542	Information	[Instrument Response] 16 ImageBlock 700 0 238 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:30.542	Information	[Instrument Response] 16 ImageBlock 800 0 238 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:30.542	Information	Transferring image block at (0,238) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:27:30.573	Information	Transferring image block at (0,238) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:27:30.636	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/14/23 13:27:30.636	Information	[Instrument Response] 16 Progress 88% 30 seconds	GMT -5:00
12/14/23 13:27:31.589	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/14/23 13:27:31.636	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/14/23 13:27:33.593	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:27:36.623	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         128 V  	GMT -5:00
12/14/23 13:27:36.670	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:27:36.670	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:27:36.670	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:27:36.670	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/14/23 13:27:37.623	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/14/23 13:27:37.686	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/14/23 13:27:38.623	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:27:40.389	Information	[Instrument Response] 16 Progress 92% 20 seconds	GMT -5:00
12/14/23 13:27:44.079	Information	[Instrument Response] 16 ImageBlock 700 0 255 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:44.079	Information	[Instrument Response] 16 ImageBlock 800 0 255 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:44.079	Information	Transferring image block at (0,255) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:27:44.110	Information	Transferring image block at (0,255) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:27:49.892	Information	[Instrument Response] 16 Progress 97% 10 seconds	GMT -5:00
12/14/23 13:27:50.673	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:27:50.736	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/14/23 13:27:50.736	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:27:50.736	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:27:51.673	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/14/23 13:27:57.610	Information	[Instrument Response] 16 ImageBlock 700 0 272 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:57.610	Information	[Instrument Response] 16 ImageBlock 800 0 272 243 17 243 290 8262	GMT -5:00
12/14/23 13:27:57.625	Information	Transferring image block at (0,272) with size 243x17 to the 700 channel.	GMT -5:00
12/14/23 13:27:57.641	Information	Transferring image block at (0,272) with size 243x17 to the 800 channel.	GMT -5:00
12/14/23 13:27:58.704	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/14/23 13:27:58.766	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/14/23 13:27:58.766	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/14/23 13:27:58.766	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/14/23 13:27:59.703	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:28:00.328	Information	[Instrument Response] 16 Progress 100% 0 seconds	GMT -5:00
12/14/23 13:28:00.719	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/14/23 13:28:00.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         120 V  	GMT -5:00
12/14/23 13:28:00.765	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/14/23 13:28:00.765	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/14/23 13:28:01.718	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/14/23 13:28:01.765	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/14/23 13:28:02.328	Information	[Instrument Response] 16 ImageBlock 700 0 289 243 1 243 290 486	GMT -5:00
12/14/23 13:28:02.390	Information	[Instrument Response] 16 ImageBlock 800 0 289 243 1 243 290 486	GMT -5:00
12/14/23 13:28:02.390	Information	[Instrument Response] 16 Status 0 COMPLETE SCAN 236.156 seconds	GMT -5:00
12/14/23 13:28:02.390	Information	Transferring image block at (0,289) with size 243x1 to the 700 channel.	GMT -5:00
12/14/23 13:28:02.390	Information	[Instrument Response] 16 Status 0 Averages (sec): setup=2.00005 line=0.338199 => total=199	GMT -5:00
12/14/23 13:28:02.421	Information	Transferring image block at (0,289) with size 243x1 to the 800 channel.	GMT -5:00
12/14/23 13:28:02.437	Information	[Instrument Response] 16 Status 0 Comparisons: setup=1.00997 line=0.927033	GMT -5:00
12/14/23 13:28:02.484	Information	[Instrument Response] 16 Status 0 Finalize motors	GMT -5:00
12/14/23 13:28:02.484	Information	[Instrument Response] 16 Progress 100% Scan complete	GMT -5:00
12/14/23 13:28:02.484	Information	[Instrument Response] 16 Result 0 OK	GMT -5:00
12/14/23 13:28:02.484	Information	Acquisition 0000004_01 finished with channels: 700 800.	GMT -5:00
12/14/23 13:28:02.671	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/14/23 13:28:02.671	Information	Scan completed successfully	GMT -5:00
12/14/23 13:28:02.781	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         123 V  	GMT -5:00
12/14/23 13:28:02.796	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          77 V  	GMT -5:00
12/14/23 13:28:02.796	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/14/23 13:28:02.796	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/14/23 13:28:02.796	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/14/23 13:28:03.730	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          74 V  	GMT -5:00
12/14/23 13:28:03.789	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          46 V  	GMT -5:00
12/14/23 13:28:04.730	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          45 V  	GMT -5:00
12/14/23 13:28:04.780	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          29 V  	GMT -5:00
12/14/23 13:28:05.739	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          29 V  	GMT -5:00
12/14/23 13:28:05.799	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          19 V  	GMT -5:00
12/14/23 13:28:06.744	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          20 V  	GMT -5:00
12/14/23 13:28:06.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          12 V  	GMT -5:00
12/14/23 13:28:07.747	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          13 V  	GMT -5:00
12/14/23 13:28:07.800	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/14/23 13:28:08.749	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/14/23 13:28:08.802	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/14/23 13:28:09.750	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/14/23 13:28:10.751	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:28:11.752	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/14/23 13:28:11.806	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/14/23 13:28:15.775	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:28:16.778	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/14/23 13:29:57.000	Information	1 image was exported to the zip file E:\Acquisitions_2023-12-14.zip.	GMT -5:00
12/14/23 13:32:46.707	Information	Successfully saved C:\Users\wmao2\2023-12-14\Image_0000001_01.tif	GMT -5:00
12/14/23 13:33:04.179	Information	Successfully saved C:\Users\wmao2\2023-12-14\Image_0000002_01.tif	GMT -5:00
12/14/23 13:33:16.514	Information	Successfully saved C:\Users\wmao2\2023-12-14\Image_0000003_01.tif	GMT -5:00
12/14/23 13:33:26.368	Information	Successfully saved C:\Users\wmao2\2023-12-14\Image_0000004_01.tif	GMT -5:00
12/14/23 13:33:34.660	Information	[Instrument Request]  17 Stop 	GMT -5:00
12/14/23 13:33:36.941	Information	[Instrument Request]  18 Stop 	GMT -5:00
12/14/23 13:33:36.941	Information	[Instrument Response] 17 Result 0 OK	GMT -5:00
12/14/23 13:33:36.941	Information	[Instrument Response] 18 Result 0 OK	GMT -5:00
12/14/23 13:33:37.722	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
